status.i4.v = hp54500_set_channel(channel.i4.v, dev_address.i4.v)
Allows user to view and change settings to an HP54504A scope
channel settings. The volts/div, channel offset, range and probe
settings are changable form these windows.
channel The scope channel that is needed:
HP_CHAN1 -> channel 1
HP_CHAN2 -> channel 2
HP_CHAN3 -> channel 3
HP_CHAN4 -> channel 4
dev_address address of scope; A previos call to gpib_select()
will have to be made to establish communication
to correct VME.
This function returns ACNET status values as follows:
CBS_INVARG invalid device address
otherwise GAS error
This function requires the following include files:
hp54500_change_dir, hp54500_display_wmem, hp54500_format_disk,
hp54500_get_disk_dir, hp54500_get_waveform, hp54500_init,
hp54500_load_from_disk, hp54500_message, hp54500_num_points,
hp54500_set_acquire, hp54500_set_channel, hp54500_set_display,
hp54500_set_root, hp54500_set_timebase, hp54500_set_trigger,
hp54500_set_utilities, hp54500_set_waveform, hp54500_store_wmem,
hp54500_write_to_disk, hp54500_num_screens, hp54500_del_disk_file.